We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||13 March 2013|
|PDF File Size:||14.1 Mb|
|ePub File Size:||3.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fail Neg Stabilize Replacing The Diasafe Plus Filter Table of contents Troubleshooting If alarm is not cleared, take the machine out of service and alert a qualified technician. K Dial Valve Failure 1 Electronic self-test failure. Infections or pyrogen reactions may also result from contamination of the extracorporeal circuit or inadequate procedures used to reuse dialyzers.
If condition persists, reduce the blood flow rate. About This Manual This manual is organized to systematically guide a patient-care specialist through the set up, 208k, and clean up of the K in daily use.
Fresenius Medical Care 2008K Manuals
To print the manual completely, please, download it. The calculated, urea-distribution, fluid volume of the patient.
This manual is organized to systematically guide a patient-care specialist through the set up, operation, and clean up of the K in daily use. If alarm does not clear, take the machine out of service and alert a qualified technician.
Page 38 Secure the patient end of the venous line over the priming bucket.
Section 18a – Diagnostic Valve Leak Tests Page 94 The result will be more accurate if both tests are done fresrnius the same blood flow rate. Using A Central Venous Catheter Assembly, installation, adjustment, or repair is to be performed only by persons authorized by the facility medical director or by Fresenius USA.
Check for residual disinfectant before using the disinfected containers.
New Fresenius k semi & annual pm manual – Topic
Acid Cleaning Frfsenius Cleaning The K should undergo an acid cleaning daily when using bicarbonate concentrates during dialysis. Deaeration And Loading Pressure Calibration Sensor Calibration Procedures In some cases, the operator must reset them. Level Detector Calibration Home Screen Settings The Fresenius K Dialysate Delivery Machine was designed and built to comply with these product specifications.
Using cuff tubing longer than 10 feet may result in erroneous blood pressure readings. Table of contents Table Of Contents The battery cartridge will pop forward. Blood Pump Calibration Verify that the pH is normal and that the conductivity is reasonably close to the theoretical value.
Page Reverse Osmosis Dial Valve Failure 2 Electronic self-test failure.
Add to My Favorites. The Control Panel Chapter 1—Overview The Control Panel The control panel see Figure 3 is located at the top, front of mwnual K machine and contains key pads that allow the user to control the operation of the K machine. K Venous Pressure Alarm Low pressure detected in the venous Blood 1 Check venous tubing for kinked line, clotting or clamps.
Comments to this Manuals Your Name. If alarm is not cleared, take the machine out of service. K High Flow Error Possible balancing chamber problem. Place the dialysate supply lines on the shunt 2.
Conductivity Cells Calibration The machine can accommodate most standard blood tubing that have pump segments ranging from 2 to 10 mm internal diameter. Continuous Renal Replacement Therapy optional Arterial, Venous And Transmembrane Pressure Fresenius Medical Care Category: Observe the system for fluid leaks and discontinue treatment if you are unable to correct any fluid leak quickly. On Line Clearance olc Problems Line from venous drip chamber is connected to a transducer protector port P and attached here to provide venous blood pressure readings.
Fresenius Medical Care k Technical Manual pages. If the external transducer protector and the internal transducer should be become contaminated with blood, the transducer protectors must be replaced, and the transducer and associated parts must be disinfected or replaced.
FRESENIUS MEDICAL CARE K OPERATOR’S MANUAL Pdf Download.
Flow Pressure Calibration After a long power down, however, only the Both Tests button manul enabled. Auto Heat Disinfect Page TP that is to be replaced with each treatment.
K Less than minimum value Entered parameter is smaller than Dialog Verify that the minimum value is acceptable. Page Other Sodium